Call us: +92 345 8515060
- Eye Instruments
- Trays & Pliers
- Plastic Surgery
- Aqua Tools
- Beauty Care
Instruments CareTreatment of brand new Instruments
Brand new instruments must be cleaned before the first sterilization or use. Protective caps and protective films must be removed completely, for example in the case of chisels, respirators or micro-surgical instruments
1- For machine cleaning, place instrument in a wire basket suitable for the cleaning process (make sure
cleaning solution and rinse water from machine comes into contact with all parts of instrument)
2- Preferably dry disposal.
3- Microsurgical instruments and instruments with fine working ends should not be treated with ultrsound.
They should be cleaned either manually or in the washing machine. In case of cleaning in a machine
please use a suitable storage rack.
1- Place instrument into a suitable disinfectant with active cleaning properties so that all surfaces, inner
cavities, lumens and openings come into contact with the solution.
2- After chemical disinfection, always rinse thoroughly with clear, flowing water.
3- Remove any dirt still clinging to the instrument with a soft synthetic brush. Do not use a scouring or
4- Clean any lumens and conduits with soft, round, synthetic brushes. Please note, the lumen and the
brush must have the same diameter.
5- Final rinsing to be done with distilled or deionized water.
6- Dry instrument with an absorbent, soft and lint-free cloth.
7- Dry lumens and conduits with compressed air.
1- Select the program according to the material (e.g. stainless instrument steel, aluminium) of the
instrument to be cleaned.
2- Final rinsing to be done with deionized water.
3- Leave sufficient time for drying.
4- Remove the instrument from the machine immediately after the program is done.
1- Let instrument cool down to room temperature.
2- Lubricate movable parts (e.g. joints and ends) slightly with special sterilization-capable, vapor-permeable
3- After each cleaning, disinfection process, check the instrument to make sure it is clean, it functions
properly and it has not suffered any damage, e.g. bent, broken, factured or worn parts.
4- Discard any damaged and defective instruments and replace.
1- Steam sterilization must be carried out according to a validated stream sterilization process.
2- When using the fractionnal vacuum method, use the 134Â°C/2-bar program for sterilization with a
minimum hold time of 5 minutes.
1- Lubricate all instruments which have any "metal to metal" action such as scissors,hemostats, needle
holders, self-retaining retractors, etc.
2- Put instruments up for autoclaving either individually or in sets.
If instruments are to be stored, let them air dry and store them in a clean and dry environment.
- SIMRIX SURGICAL CO. Head Marala Road, Chowk Gohadpur, Sialkot 51310 Pakistan.
- +92 52 4564188
SIMRIX SURGICAL CO. © 2020